FAQ’s for Oricom Babysense7
Q. Is it a medical device?
A. Babysense7 is included as a medical device on the Australian Register of Therapeutic goods (ARTG No. 97479).
Q. Is it simple to set up and easy to use?
A. Yes, extremely simple to set up and operate, however we strongly recommend the user guide instructions be followed and functionality regularly tested.
Q. Is it portable?
A. Yes, it uses 4xAA Alkaline batteries (not supplied) so you can move it easily.
Q. I’ve read that the babysense7 monitor runs on batteries. How long (on average) will the batteries normally last before needing to change them?
A. The babysense7 is Powered by 4 x AA, 1.5-volt Alkaline batteries (not supplied). Power consumption is approximately 1mA. Battery life is approximately 6 months using high quality alkaline batteries.
Q. Can it be used in cribs, bassinets, travel cot?
A. Yes, as long as the base is rigid and a suitable mattress is used and you must ensure that excess cords are secured under the mattress out of reach of baby or other infants. Modifications may be required to the bassinet for sensor pad wiring access and controller unit placement. For further advice please consult an Oricom Representative.
Q. How many sensor pads does it have?
A. It comes with two highly sensitive sensor pads that cover the area of a standard sized crib or cot.
Q. What age groups is it suitable for?
A. It is suitable for healthy infants up to one year (12 months) of age (it cannot be used for children or adults).
Q. If I have twins, can I use with both babies in one cot?
A. No you will need to have babies in separate cots, each with its own Babysense7 monitor.
Q. Lights on Control unit do not illuminate when turned ON?
A. Batteries are not inserted incorrectly or may be flat. Check batteries are inserted in the Control unit observing the correct polarity. Control unit requires 4 x AA 1.5V Alkaline batteries (not supplied).
Q. Low battery voltage warning (red light) on the Control Unit is flashing?
A. Batteries are low, replace all 4 batteries.
Q. Alarm goes off after a few hours
A. The mattress is not touching the sensor pads.
The mattress is raised (there is an object near the sensor pads).
Two sensor pads are connected but the baby is not yet crawling.
Only one sensor pad is connected and the baby has crawled off the sensor pad.
The mattress does not enable adequate transfer of baby movement – The compositions of some cot mattresses may not conduct vibrations associated with babies breathing movement and may not be suitable with Babaysense7.
Do not use Babysense7 with a memory foam mattress.
Make sure that the mattress and sensor pads are touching.
Make sure that there are no objects next to the sensor pads.
Connect sensor pads as per the user manual. If baby is not yet able to crawl, use only one sensor pad – and make sure sensor is placed underneath baby’s location in bed to ensure greater relative sensitivity.
Place a sheet of plywood (not supplied) under the sensor pad so that the sensor pad and the mattress will be tightly pressed together.
Q. Delayed alarms while taking the baby out of the bed
A. Performance test is not correct.
Do not conduct performance test by taking the baby out of bed and waiting for the alarm to go off. Kindly conduct performance test according to the performance test instruction in user guide.
The alarm does not go off within 20 seconds when performing the correct performance test
Environmental Influences detected by the sensor pads, such as wind, fan, air-conditioner, toy mobile vibrations.
Perform a test which will detect the source of the movement that the device detects, eg; sources of wind such as air conditioner, an open window or a fan, movements of baby mobile (which can vibrate even if it’s not turned on) walking around the bed area or an intercom placed near the bed with intense transmission levels. Eliminate sources of wind, remove the mobile etc. After neutralising all environmental interruptions, perform the test again.
Q: Can I use the Babysense7 on my 13-month-old infant?
A: Babysense7 is intended for healthy newborn infants from birth and up to 1 year of age.
Babysense7 is set to recognise a defined breathing movement rate of infants within this age group (the alarm sounds if breathing stops for 20 seconds or if the breathing rate slows to below 10 breaths per minute).
Q. Can I use Babysense7 if my infant needs constant medical supervision?
A. No. You should contact your GP or medical specialist and seek their advice on what alternative suitable methods of monitoring your infant are available.
Q. Why is it recommended that only one pad be used in a bassinette?
A: The Babysense7 detects the movement produced by a baby’s breathing. A newborn, being smaller, may not have as much movement than a 3 or 4-month-old baby, and when two sensor pads are connected it does reduce the sensitivity of the pads, a little, no problem for an older baby but maybe a little more difficult to detect the movement for a new born baby. This is why it is always best to follow the instructions in the user guide including performing a regular (daily) test of the monitor.
Babysense7 is intended for use as an Infant Apnoea Alarm.
ALWAYS READ THE LABEL AND USER GUIDE, USE ONLY AS DIRECTED.
Q: My SC720 is displaying Lost Link on the parent Unit. How do I fix this?
A: Move the parent unit closer to the baby unit.
Try resetting your monitor to avoid busy channels:
1. Turn both camera and parent units off and remove the battery in the parent unit. Leave unplugged for 1 minute.
2. If you have Wi-Fi in the home, make it busy by streaming a video on YouTube or similar.
3. Turn the camera back on wait for 30 seconds.
4. Plug the battery into the parent unit and turn on.
If the display is still showing [LOST LINK], please re-register the units as instructed on page 7. When re-registering it is best to keep the camera and parent units close together (within 1m) and keep clear of all other electronic devices including mobile phones, tablets or computers.
Q: How do I register a new camera unit?
A: The parent and baby unit are pre-registered. If needed, you can register up to 4 baby units (cameras) to the one parent unit.
To re-register your baby unit, or register a new baby unit, follow the procedure below:
(1) Switch on the parent unit.
(2) Press and hold the camera button on the parent unit until registration icon appears on the screen.
(3) After the registration icon appears, press or to select the camera channel [C1] or [C2] or [C3] or [C4] that you want the camera unit to pair with.
(4) Press and release the menu key to confirm camera number registration.
(5) Connect camera to mains power. Press the registration button once (at the back of the camera unit).
(6) If registration is successful, camera video appears on parent unit screen.
(7) If the registration time is expired or registration fails, a cancel icon appears on the LCD.
NOTE: When a channel has a registered camera, new registration on this channel will override the existing registration.
Q: How many cameras can I connect to the one parent unit?
A: Up to four cameras can be connected to the one parent unit.
Q: The amber power LED is flashing?
A: The battery level is low. Return the parent unit to the charger.
Q: My parent unit screen goes blank after 10 seconds.
A: VOX function is activated. Turn off VOX if required.
Q: Why is my battery not charging?
A: Check the battery is connected to the battery connection in the battery compartment. After a period of time, the battery may need to be replaced due to everyday usage. Purchase additional batteries from Oricom website.
Q: My screen is showing in black and white, not colour. What has happened?
A: Night function is active. Increase the light level in the baby’s room if required.
Q: What are the lullabies on this unit?
10 Minuet In G Major
Q: What is the Scan Mode?
A: When multiple cameras are registered to the parent unit, the scan mode will cycle through each camera, displaying the image for 10 seconds. The audio for the associated image will be played, i.e. if camera 1 image is displayed, you will receive the audio for camera 1. The parent unit
will continue to scan through all cameras connected. If you need to view a camera for more than 10 seconds, deactivate the scan mode.